Vision and Mission
Quality & Environment
Service & Support
Service & Support
Weighing & Automation
NIR, Optical and Imaging
Service & Support
Service & Support
Mobile loading cells
Which is the degree of AgriNIR’s accuracy and why are frequent measurement on Dry matter necessary?
analysis equipment has a 2% accuracy on moisture and 3% on all the other parameters (STARCH, CRUDE PROTEIN, ASH, ADF and NDF) while traditional laboratory analysis equipment can guarantee an accuracy that is 1,5% for moisture and 2% for all the other parameters.
Why it is possible to introduce a new forage category as requested by the client?
Based on our experience, if we have available 100 samples representative of the variability present on the territory (well distributed in the area where the AgriNIR
is used) we can develop a stable calibration.
Are the formulas for the parameters sugar, phosphorus and potassium loadable on the AgriNIR (TM) software?
We can create calibration curves for all parameters that can be mathematically calculated on the basis of the existing nutrients predicted by AgriNIR
When making the single analysis (not multiple) with the sample (without even taking it from the box and putting back), predictions are different, deviation is within 5% - why is it so?
Regarding the deviation that has been recognized performing single analysis on samples, it is quite normal in case of material not homogeneous (as it is for example hay).
How to manage the sample cup while do you perform analysis?
In order to obtain accurate and repeatable predictions by AgriNIR
machine it is important to take the sample’s cup outside from the AgriNIR
machine between analysis.
Which languages does the DTM (TM) support?
is a multi language software so it provides you the possibility to work with several languages.
Is it possible to create customizable report on demand?
gives you the possibility to have the report about the recipes and the loading program or about the execution of the recipes or unloading program you executed on the Scale mounted on the Feed Mixer.
How are the manual weighing managed?
The Scale, which is installed on the Feed mixer, gives you the possibility to save on the memory card (if it is connected) the actual weight of the feed, actually present on the feed mixer.
How are ingredient and cost managed?
gives you the possibility to define the costs for each ingriedent, so when you create a recipe you will know the total cost related to each ingriedent and the total cost for the recipe.
Is it possible to associate one group to one recipe?
gives you the possibility to insert all the groups of your animals and you can link one group to one or more than one recipe.
How to manage the loading program?
gives you the possibility to define two kinds of recipes:
How to manage the unloading program?
gives you the possibility to define the unloading program, so after loading the feed on feed mixer, you can decide how to discharge the feed.
Is it possible to store the load and unload "Execution"?
gives you to show all the executions stored by the dinamica generale Weighing System; as a matter of fact you can transfer the recipes and the unloading program into a memory card.
Is it possible to get the execution data from the Scale?
If the memory card is connected to the Scale, the weight box saves a lot of details as the effective loaded quantity for each component.
How can I track the costs?
Consulting the details of your executions, you can see that the DTM
shows you the details about the Programmed costs and the Effective costs.
Does the DTM (TM) notices the operator error?
The operation of loading and unloading are executed by an operator that drives the feed mixer.
Is it possible to register the operator and track the driving?
In this case you can define an account related to each operator, that will be transferred on the memory card.
Wich are the steps to be carried out if the DTM (TM) does not recognize the card?
You must check if the COM port is correct in the hardware management and you must plug and unplug the card.
Is it possible to print the information coming from the indicator of my feed-mixer on DG printer? Is it necessary some other device (adapter, ...) in order to obtain this?
There are two possible ways for printing the information and data during or after the weighing sessions:
Why buy the mobile load cells instead of the shear beam industrial?
Because they provide a strong and accurate load sensing device at reasonable costs; the robustness of the mobile cells is ideal for mobile applications, such as feed mixers, grain carts or others
Why vehicles weighing in motion?
Dina3 Axle Scales are mobile, reliable and high-capacity weighing devices.
What is the accuracy of the Dina3 "Checker/OIML" Systems?
The accuracy is +/-1% of the vehicle weight in dynamic mode and +/-0,5% of the vehicle's weight in static mode. Remember that the accuracy is depending on the site construction. It's strongly recommended to follow DG's instructions to build the site casting.
Which type of vehicles can I weigh?
With the inground installation or extra-leveller you can weigh any kind of vehicles. By using the ramps you can basically weigh two axles vehicles.
What is the maximum allowed speed on the platforms?
The suggested speed is 1-3km/h.
Can I set a client code?
Client code can be set through the CMD key. A Tare value is associated to a client code and stored in memory.
How can I export weighing data?
The indicator can be directly connected to a PC through serial port;
Are the weighing data stored in the indicator?
All the weighing session are stored in the "Weighing History" (CMD key) and can be recalled at any time.
How must the system be calibrated during the installation?
Platforms are calibrated in our factory before delivery.
How can I change the language on the display?
Press "Enter" during the indicator's start up. In this way, you will enter "System Configuration" menu.
What happens if "ERR02-Too Few Data" is displayed?
It means that the movement of the vehicle was not constant on the platforms.
What happens if the message "Wrong Platform identification" is displayed?
Check that both cables are connected properly to the platforms and to the indicator.
What happens if the message "Platforms Cables Inverted" is displayed?
Switch the unit off, invert the platforms cables and switch the unit on again.
What happens if the message "Zero Setting Error" is displayed?
In this case the condition of the indicator is "Locked". Switch off the power unit and verify that:
What if the printer doesn't print?
Verify that the printer is correctly connected to the indicator. Check out if the printer is loaded with thermal paper.
DTM (TM) SW
CONTACT DINAMICA GENERALE
We will be pleased to answer any questions you may have
Term of use
Dinamica Generale S.p.A.
Via Mondadori, 15 - 46025 - Poggio Rusco (MN)
company registration number 01599500202
REA MN 169392
Paid-in capital 1.000.000,00 I.V.
Copyright © 2017-2020
. By continuing to use this website, you consent to our use of these cookies.